On A function generator, using RS RS , binary download of arbitrary waveform does not always work. Consider PXI Arbitrary Function Generators NI PXI Waveform Generators can produce precise waveforms including sine, square, triangle, and ramp as well as arbitrary, user-defined waveforms by using data sequences or by streaming continuously from a host or peer-to-peer instrument within the PXI system. For this example, focus on the section between the Configure Arbitrary Waveform node and the Enable Output node. Back to Top 5. With some more tinkering, I’ve discovered 2 things. Accepted by topic author jto2. Customize the panel of the VI to fit your application needs.
|Date Added:||22 December 2017|
|File Size:||16.23 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Drop all the indicators and controls from the Unplaced items box on the panel. First of all, that wait seems really weird to me.
If the waveform is large and you are using RS interface, binary data format is not supported. Select the number of bursts you want the function generator to produce using the burst count control.
Keysight A signal generator
When writing an instrument control application for a serial instrument, you communicate with a COM port rather than the serial instrument connected to the COM port. Install Agilent Series Instrument Driver NI labviwe and supports thousands of instrument drivers for third party instruments to help engineers and scientists llabview measurements. Hundreds of built-in functions are included with LabVIEW and the Agilent 33XXX series driver After placing the Configure Burst node, be sure to connect the purple instrument wire and the yellow error wire appropriately.
Most Active Software Boards: Thanks for the help! Message 6 of 7.
Keysight Technologies / Agilent Technologies 33210A
Add an input to change the number of bursts you want the instrument to perform by right clicking the burst count terminal on the albview side lsbview the Configure Burst node and selecting Change to control. When performing frequency measurements of frequenciesi tie the time bases of both the standards and the counters together dependi TestStand, a ready-to-run test management software, is designed to help you develop, execute, and deploy automated test and validation systems faster.
Press the Offset key and hold it do Now the waveform changes amplitude without turning off the output. Using Agilent 33xxx example: If you select the device, labviiew can see additional properties, such as the device name, and any software or drivers that are installed on your system to support this device.
NI develops and supports thousands of instrument drivers for third party instruments to help engineers and scientists labvview measurements. The instrument is now configured. Has anyone else had any problems with this pulse gen, in that if it sits for awhile, front keyboard commands are sluggish to work.
The mark LabWindows is used under a license from Microsoft Corporation. However, it is still showing twice the set voltage.
Keysight 33210A Signal Generator, 10 MHz, 1 Channel
Most function generators have a built-in dc offset that can be added to any waveform. With it you can leave de output on and change the waveform, amplitude, frequency, etc Thanks last modified by shaobing I’ve also tried factory reset as well. Thanks for the input. You can connect your Agilent series function generator to your host PC using a method that works for your instrument. You can also troubleshoot and make manual function calls to your instrument to ensure that it is working as expected before moving into automation.
The recommended calibration interval for these devices is 12 months.